Here is the abstract you requested from the IMAPS_2012 technical program page. This is the original abstract submitted by the author. Any changes to the technical content of the final manuscript published by IMAPS or the presentation that is given during the event is done by the author, not IMAPS.
|A CERAMIC TUBULAR PROBE FOR ONLINE SUBSTANCE CONCENTRATION MEASUREMENT, MANUFACTURED BY CERAMIC INJECTION MOLDING|
|Keywords: three-dimensional ceramic tubular probe, online substance concentration measurement, ceramic injection molding|
|The current research presents recent respective to the work development of a ceramic tubular probe for online substance concentration measurements. The aim was to develop a robust and acid-resistant sensor device, which can be easily included in existing procedural pipeline systems. To archive those goals a lot of factors had to be checked. For the substance concentration measurements a capacitive sensor effect was chosen. With this method even low substance concentrations down to one-tenth of a per cent can be indentified. For the package material zirconium oxide (tetragonal zirconia polycrystal " TZP) was used. Zirconium oxide is a technical ceramic which is wear-resistant, acid-resistant, has a low thermal conductivity, is electrically isolating and can be uses in a ceramic injection molding (CIM) process. In the phase of the sensor design process multiple geometries for the sensor effect and integration space for the evaluation electronics had to be considered. A standardized DN 10 DIN 32676 flanged joint was also added for an unproblematic connection to the pipelines. All these needed geometries had to be integrated into one ceramic element. As a result of these requirements a 3D CAD model of the sensor element was designed. The CAD-file has shown that there was only the CIM technology left to comprehend developed sensor geometry. CIM is a low cost process for large-scale production which is distinguished by high size accuracy. In the CIM process the material shrinkage, this is caused by the needed debindering and sintering steps, had to be considered. The developed ceramic tubular probe was successfully tested in multiple fluidic systems. It has left the test phase and is now ready for maturity phase.|
|Matthias Hartmann, PhD Student
Otto-von-Guericke University of Magdeburg